VacuDAPVacuDAP VacuDAP is a Dose Area Product Meter used for patient dose measurements.
Distributed only in North and South America by Radcal Corporation.
Radcal is proud to announce our newly formed partnership with VacuTec, Dresden, Germany, to bring the VacuDAP meters to North America. “VacuTec is a very well known and respected manufacturer throughout Europe” states Ken Mettler, President, Radcal, “and we are happy to be a provider of their DAP meters for the North American market.” The Dose Area Product meters, VacuDAP, are used for patient dose measurements. It does not affect the routine operation and gives the opportunity to measure total dose to the patient site in order to minimize the radiation risk.
In order to comply with FDA regulation 21 CFR 1020.32 for fluoroscopic equipment, Radcal will initially offer the Air Kerma meter, VacuDAP Duo. The two-field measuring system is designed for simultaneous measurement of air kerma, air kerma rate, DAP, DAP rate and irradiation time. The measuring system works via serial interface with the host PC or a stand-alone display unit. Bluetooth versions are also available.
- Key Features and Benefits
Full Selection of Ion Chambers Wide range of ionization chambers and configurations for most of medical diagnostic x-ray equipment for various applications. Wireless Connectivity Bluetooth wireless technology available for the system. Standards Compliant Measures air kerma and air kerma rate in accordance to the standards IEC 60 601-2-54, IEC 60 601-2-43 and 21 CFR 1020.32
- For expert advice on your exact application please call us at 1-626-357-7921 ext 111 or email email@example.com for the product recommendation that would best suit your needs.
GENERAL: Quality equivalent filtration (70 kV) 0.2 mm Al Light Transparency(rectangular chambers) >70% Radiation quality (40 ... 150) kV Atmospheric pressure (80 ... 106) kPa Temperature (+10 ... +40) °C Air humidity (10 ... 80)% rel. humid. (max. 20 g/m3) DIMENSIONS: Rectangular transparent measuring chambers: Active area 123 mm x 123 mm, 147 mm x 147 mm CIRCULAR MEASURING CHAMBERS: Active area / outer dimension (diam.) 44 mm/60 mm, 72 mm/100 mm(non transparent), 68 mm/90 mm (transparent) ELECTRONICS FOR CIRCULAR CHAMBERS: 80 mm x 50 mm x 17 mm DISPLAY UNIT: 160 mm x 94 mm x 37 mm DAP: Digital resolution 0.01 µGy·m2 Measuring range 0.1 ... 99 999 999 µGy·m2 DAP RATE: Digital resolution 0.6 µGy·m2/min Mesuring range 6 ... 1 800 000 µGy·m2/min Useable active area 1 ... 200 cm2 DOSE*: Digital resolution 0.003 mGy Measuring range (0.03 ... 99 999 999) mGy DOSE RATE*: Digital resolution 0.18 mGy/min Measuring range (1.8 ... 17 000) mGy/min Useable active area (2 ... 200) cm2 Minimal field width 1.4 cm
* Distance focus-chamber: 28 cm; Distance focus-reference point: 100 cm*Specifications subject to change.